Octane Meter SHATOX SX-200
1.1. The Operations manual for OCTANE METER SHATOX SX-200 petroleum products quality
analyzer (hereinafter called OCTANE METER) is intended for studying the device and contains description of
its design, the principle of operation, performance specification, and establishes the operating rules to guarantee
trouble-free work of the device.
1.2. It is not required any special training to operate Octane Meter. Nevertheless, we recommend you to
study the present Operations manual carefully before starting operations of the device.
2. Description and operation of Octane Meter
2.1.1. Octane Meter is designed for determining the following:
· motor petrol octane levels by motor and research methods;
· diesel fuel cetane rating. In addition, it is possible to determine diesel fuel chilling temperature;
· motor oil purification efficiency and oil products dielectric capacitivity;
· puncture potential and transformer oils dielectric dissipation.
The results of measurements are displayed on the screen.
2.1.2. Octane Meter is manufactured as a portable small-sized device and is intended for rapid petroleum
products quality monitoring in field and laboratory conditions. Operating conditions: air temperature in
the range from – 10 °С up to + 45 °С.
2.1.3. The device is supplied with power from 4 elements of АА type or from an external direct current
power supplier with voltage of 6 … 7 V.
Technical specifications of Octane Tester SHATOX SX-200
|Measurement range of octane numbers (ON), ON units||40-120|
|Acceptable limit of octane number measurement basic error, ON units (max)||0.5|
|Limit of acceptable difference between parallel octane number measurements, ON units (max)||±0.2|
|Measurement range of cetane numbers (CN), CN units||20-100|
|Acceptable limit of cetane number measurement error, CN units (max)||±1.0|
|Limit of acceptable difference between parallel cetane number measurements, CN units (max)||± 0.5|
|Motor oil purity level measurement range, % max||95|
|Acceptable limit of motor oil purity level measurement error, % max||0.1|
|Limit of acceptable difference between parallel motor oil purity level measurements, % max||0.01|
|Fuel and oil dielectric conductivity measurement range, units||01-May|
|Acceptable limit of dielectric conductivity measurement error, units (max)||0.001|
|Limit of acceptable difference between parallel dielectric conductivity measurements, units (max)||0.001|
|Transformer oil disruption voltage measurement range, kV||5-100|
|Acceptable limit of transformer oil disruption voltage measurement error, kV (max)||1|
|Limit of acceptable difference between parallel transformer oil disruption voltage measurements, V (max)||0.2|
|Transformer oil dielectric dissipation measurement range, %||0.01-5|
|Acceptable limit of transformer oil dielectric dissipation measurement error, %||0.01|
|Limit of acceptable difference between parallel transformer oil dielectric dissipation measurements, units (max)||0.001|
|Measurement time, s||01-May|
|Threshold of low battery voltage indication, V||5.4|
|Sensor, mm||60 х 100|
|Electronic unit, mm||100 х 210 х 25|
|Mass of octanemeter, gr||680|
|Mean-time-between-failures, hour (min)||1000|
Returns and Exchanges
There are a few important things to keep in mind when returning a product you purchased.You can return unwanted items by post within 7 working days of receipt of your goods.
- You have 14 calendar days to return an item from the date you received it.
- Only items that have been purchased directly from Us.
- Please ensure that the item you are returning is repackaged with all elements.
Ship your item back to Us
Firstly Print and return this Returns Form to:
30 South Park Avenue, San Francisco, CA 94108, USA
Please remember to ensure that the item you are returning is repackaged with all elements.
For more information, view our full Returns and Exchanges information.